Intensive Care Ventilator ZISLINE MV300
Evaluation of patients metabolic needs
Cardiac output by Fick method
Modes of spontaneous breathing
Non-invasive ventilation modes
|Rate of breathing
0–100 H₂O (mbar)
0.5–20 cmH₂O (mbar)
|Positive end-expiratory pressure
0–50 cmH₂O (mbar)
100–250 V, 50/60 Hz
|Input oxygen pressure
0.15–0.6 MPa (1.5–6 bar). It is allowed to use low-pressure oxygen sources with operating pressure range: 0.05–0.15 MPa (0.5–1.5 bar)
Visual and sound alarm
Ethernet for connection to PC, USB
|Maximum (peak) flow on inspiration
- 01 /
- 02 /
- 03 /
- 04 /
This monitoring method is recommended for use in intensive care units and operating rooms to improve patient safety.
Capnography allows to assess the endotracheal tube location, the resuscitation effectiveness. This type of monitoring is necessary for patients with increased intracranial pressure.Volumetric capnometry has additional capabilities:
- allows to assess the alveolar ventilation;
- tracks the change in physiological dead space at the artificial ventilation.
esophagus. The pressure in esophagus is equal to the intrapleural pressure.
Among the main principles of protective artificial lung ventilation the PEEP is considered to be an important component for the prevention of atelectotrauma.
P transpulmonary = P alveolar — Ppleural.
Transpulmonary pressure is the only objective criterion for setting up PEEP. Its monitoring allows reducing or eliminating lung injuries during the ventilation.
Stress index is an indicator of the correct choice of PEEP and the inspiration volume Vt. Its deviation from “1” shows non-optimal choice of ventilation parameters.
|carbon dioxide excretion
|resting energy expenditure
In addition to directly measuring the actual resting energy expenditure (REE), this method calculates the respiratory quotient (RQ) — the ratio of carbon dioxide release rate to oxygen consumption rate and assess the contribution of each macronutrient to the total metabolism.
Documents and manuals
Have some questions? Write us and we answer you
Partner Registration is available for our Distributors and allow you to download our latest Technical Data Sheets, User Manuals, Service Manuals, Certificates and Marketing Materials.
Registration is an easy process. Please click the 'Register' button to begin, fill in the form, and you will be contacted by our International Sales Team within a couple of working days about your registration, User ID and password.
If you are having problems filling out this form, send us an email: email@example.com for assistance.